Dräger Fabius OS Features Familiar operation and layout ensures ease of use and a short learning curve. RS 232 enables communication with automated record keeping systems for enhanced patient throughput. Convenient compact breathing system (COSY). Ventilates any patient - even as your needs change to accommodate sicker patients with more complex procedures. Consumes no drive gas, allowing dramatically increased ventilation time and reduced cost. Dräger Fabius OS Specifications Weight: 296 lbs. (134.2 kg) (base unit without vaporizers or cylinders). Dimensions: 40.7 x 52 x 33 ″ (103.5 x 132 x 84 cm) (W X H X D) (with Cosy and 3-Vaporizer Mount). Power Supply: 100 - 240 VAC, 50/60 Hz, 2.3 A max. (non-configurable). Operation Time: Up to 120 minutes, minimum 45 minutes. (With fully charged batteries.) Ventilator E-vent: Electronically controlled, electrically driven. Operating Modes Standard: Volume Controlled Ventilation, Manual Ventilation, Spontaneous Breathing. Operating Modes Options: Pressure Controlled Ventilation, Pressure Support. Breathing frequency: 4 to 60 1/min. Max. Minute volume (MV): 0 to 99.9 L/min. Positive end-expiratory pressure: 0 - 20 cmH2O (PEEP). Inspiration / Expiration ratio: 4 : 1 to 1 : 4 (TI:TE). Pressure Limiting: 15 - 70 cmH2O (PMAX). Tidal Volume (VT): 20 - 1400 mL in Volume Control. Inspiratory pause (TIP:TI): 0 - 50 %. Inspiratory pressure (PINSP): PEEP + 5 to 65 cmH2O. Inspiratory flow (InspFlow) 1: 0 - 75 L/min in Volume and Pressure Control. 10 - 85 L/min in Pressure Support. Pressure Support Level (Δ PPS): PEEP + 3 to 20 cmH2O. Min. Frequency for apnea-ventilation: 3 - 20 1/min and “ OFF ” (Freq. Min). Trigger: 2 - 15 L/min. Integrated safety functions: Sensitive Oxygen Ratio Controller (S-ORC) guarantees a minimum O2 concentration of 23% in an O2/N2O mixture. N2O cut-off if O2 fresh gas valve is closed or if O2 flow is less than 0.2 L/min. Audible and visual (flashing red LED) indication in case O2 pressure drops below 20 psi (1.4 bar) ± 4 psi (0.3 bar). In case of electricity and battery failure, manual ventilation, gas delivery and agent delivery are possible. Positive pressure relief valve opens at 75 ± 5 cmH2O. Negative pressure relief valve opens at - 7.5 to - 9 cmH2O. Range of fresh gas flow indicators: 0.00 to 12.0 L/min. Total fresh gas flow meter: 0 to 10 L/min, calibrated with a mixture of 50 % O2 and 50 % N20 mixture. O2 flush: At 55 psi (3.8 kPa x 100): max. 50 L/min. At 50 psi (3.4 kPa x 100): max. 35 L/min. Vaporizer mount: Auto Exclusion Drä ger mount. Control screen: 6.5” (16.5cm) black/amber. Monitoring: Continuous monitoring of inspiratory O2 concentration, breathing frequency, tidal volume, minute volume, mean or plateau pressure, peak airway pressure as well as PEEP. In addition, all fresh gas flow information is displayed as virtual flow tubes. Serial interface: 1 x RS 232 (Standard). Protocols: Vitalink and Medibus. Data available for export: All fresh gas flow, ventilation and O2 data, flow and pressure real-time waveform. Volume of CO2 absorber: 1.5 Liter, Option: Dräger Medical’s prefilled CLIC absorber (1.2 l). Gas supply: O2, N2O & Air. Cylinder yokes: O2, N2O, Air, pin index yokes. Additional accessories: Gas scavenging, patient suction, power outlet strip, additional O2 flow tube. Volume of entire compact breathing system: 1.7 Liter + bag.