idoc seamlessly blends the two recording options together into one compact configuration, allowing the user to simply attach one digital component to any Marco G-model Ultra Slit Lamp. With a simple press of the joystick switch, high-quality digital images and live video streams with no time delay are instantly captured and stored into idoc's simple software program. All basic operations are conveniently displayed on idoc's main screen, allowing the user to quickly and easily access patient files.
Hardware & Software Specifications: OS Windows XP 32 bit/64 bit (All Editions) SP2/SP3 Windows Vista 32 bit/64 bit (All Editions) Windows 7 32 bit/64 bit (All Editions) SP1 CPU Intel Core i3-213Ø CPU @ 3.40GHz RAM 4GB HD 200GB or more Drive CD-RW Video Card Any video card which has enough capacity to show high quality images. (Consult your IT professional.) I/F Desktop IEEE 1394A (Firewire) PCI board is required to be installed, 1 RS232-C Serial Port (or Serial to USB adapter), 4 or more open USB 2.0 ports at the rear of the computer case. I/F Laptop Externally powered IEEE 1394A (Firewire) card for the specific laptop auxiliary card slot. These are absolute minimum requirements. Exceeding these requirements will enhance performance. Camera Specifications: Image Device ½” Type progressive scan SONY IT CCD Effective Picture Elements Up to 1388x1038 pixel Resolution Depth 8 bit Color Modes Raw8 (Mono8), YUV4:2.2, YUV4:1:1 Digital Interface IEEE1394; DCAM V1.30 Transfer Rate 100, 200, 400 Mb/s Frame Rate 15MHz at full resolution (1388 x 1038) Power Requirements DC 8V-36V via IEEE1394 Cable PowerConsumption Less than 3 Watt(@12 VDC) Beamsplitter ratio for adaptor is 70% camera and 30% examiner.